Flender Mechanical Power Transmission Couplings available from jbj Techniques 9 #DriveLineHarmony www.jbj.co.uk/couplings.html 2) Installation AIQ Core Before using the equipment for the first time, ensure that the equipment is not damaged before installation. If in doubt, you should contact a qualified electrician or jbj Techniques Limited technical office, telephone: 01737 767493 or email: email@example.com 2.1 Device Mounting If the sensor is ordered together with a FLENDER gearbox, the sensor is already mounted on the gearbox. In this case, proceed to point 2.2. When ordering the sensor for an operating gearbox, a drawing is enclosed with the device showing the best mounting position. 1. Prepare the mounting surface for installation of the device. The device should ideally be mounted above the input shaft on the gearbox housing. A system structure that is as rigid as possible should be en-sured. The ideal positioning can be viewed in the en-closed sheet "Positioning". Note: Mounting on covers is not recommended and leads to incorrect measurement results in vibration measurement. 2. Prepare threaded hole according to sketch. A reduction >Ø 25mm is recommended to compensate for unevenness on the gearbox surface. Note: Alternatively, the mounting adapter can be glued using an adhesive suitable for vibration measurements (e.g., Loctite 330). 3. Clean the attachment surface and protect it from corrosion with suitable aids. 4. Attach the mounting adapter with cylinder head bolt (M6x14), paying attention to the orientation. Note: The final cable routing is specified hereby. 5. Attach sensor interface to mounting adapter (M6x14). i !